Infusion Pumps Face Class 1 Recall after 55 Injuries, 1 Death
Alaris System Pumps recalled due to serious errors.
Monday, March 9, 2020 - Becton Dickinson CareFusion 303 is recalling 774,000 Alaris Infusion Pump Systems and its modules due to several problems that can potentially cause death in patients. The FDA has identified this as a Class I recall, the most serious type of recall.
The recalled pumps, used in hospitals and other adult, pediatric, and neonatal care settings to infuse medications and monitor vital signs, have experienced programming delays, battery alarm failures, programming failures, and other problems that can be dangerous to patients.
To-date, 55 injuries and one death have been reported due to under-infusion, over-infusion, delay of infusion, or interruption of infusion of medications.
According to the FDA recall notification the following errors have occurred:
Users utilizing modules with software version 9.33 or earlier have experienced:
Delay options programming
Keep vein open (KVO) / "End of Infusion" alarm priority
Software/System errors (System Error 255-xx-xxx)
Low battery alarm failures
Use-related errors related to custom concentration programming
Users utilizing modules with software version 12.1.0 have experienced:
Low battery alarm failures
Use errors related to custom concentration programming
Keep vein open (KVO)/ "End of Infusion" alarm priority
KVO Rate Not Available When Using Delay Options programming
All lots of the affected models of Alaris System Pumps and modules have been recalled. The following model numbers and units are included in the recall:
BD Alaris™ System PC Unit Model 8000, software versions 9.5 and prior
BD Alaris™ System PC Unit Model 8015, software versions 9.33 and prior, and software version 12.1.0
BD Alaris™ Pump Module Model 8100, software versions 9.33 and prior, and software version 12.1.0.
Alaris™ Syringe Module Model 8110, software versions 9.33 and prior, and software version 12.1.0
Alaris™ PCA Module Model 8120, software versions 9.33 and prior, and software version 12.1.0
Alaris PC units with software version 9.33 and older were distributed from July 2004 until October 31, 2019. Alaris PC units with software version 12.1.0 were distributed from December 18, 2019 until January 23, 2020.
Letters were sent to warn hospitals, providers, and consumers of problems with the pumps February 4, 2020. In the letters, Becton Dickinson told customers that the systems may be "vulnerable to the potential risk of multiple system and software errors." Consumers were directed to a pdf list of ways to mitigate some of the risks, and were told they would be notified of a software update when it becomes available.
OnderLaw, LLC in St. Louis has developed a solid reputation of excellence as a personal injury law firm handling serious injury and death claims across the country. Its mission is the pursuit of justice, no matter how complex the case or strenuous the effort. OnderLaw has represented clients throughout the United States in pharmaceutical and medical device litigation, including Pradaxa, Lexapro and Yasmin/Yaz, Actos, Risperdal and others. OnderLaw has won verdicts of more than $300 million in talcum powder ovarian cancer lawsuits and represent more Glyphosate victims than any other firm in the country.